Calibrating MD Smart Tool Level
Lost the handy-dandy instructions for calibrating my 48″ MD Smart Tool level. Anybody know of a source for these instructions can be located quickly? Need to calibrate before I put the tool through a work-out tomorrow. Thanks!
bought one last week. put the level on a flat surface with the monitor facing you. turn it on. hold the calibrate button until the number starts to flash. wait 10 secs. rotate level end for end so now you are looking at the back of the monitor. hold the calibrate button again until it reads cal 2. wait 10 secs. its calibrated. that said, i took mine back the next day. the booklet says it has to be recalibrated if the temp changes a few degrees and every couple days. i found it had to recalibrated every time i picked it up.and sometimes the reading did not match the bubble. oh-plumb calibrates like level. set it up, touch the button and rotate end for end. good luck.
Thanks, Paul. That did the trick!
Shawn @ Stolte Construction